Request a Quote

Please, provide us with some details about you or your company. We will just send you the quote, no spam or ad emails.

Primary unit you are interested in *

How many

1

50

Other unit you are interested in (optional)

How many

1

50

Your message

We will send you the quote or contact you in less than 48 hours.

Inertial module

PI-08M

PI-08M

PI-08M is our smallest inertial module for integration to your systems and designs. Essentially, it is an internal part of the IMU PI-08. It is smaller and cheaper than the IMU, not environmentally protected and has different interfaces and power consumption. It provides the same level of accuracy if it is correctly mounted, is not subjected to severe temperature gradients or exposed to powerful electro-magnetic interference. We have recommendations in the manual on how to integrate and use our modules in your systems so the best performance is achieved.

Full-scale range

Bias (In-Run Stability), (1σ)

Bias (Over Temperature), (1σ)

Angle Random Walk, (1σ)

Rate Noise Density

Scale Factor

Error over temp., (1σ)

Non-Linearity , (1σ)

Bandwidth (-3dB)

Misalignment, (1σ)

g-Sensetivity

Bias (In-Run Stability), (1σ)

Bias (Over Temperature), (1σ)

Angle Random Walk, (1σ)

Rate Noise Density

Scale Factor

Error over temp., (1σ)

Non-Linearity , (1σ)

Bandwidth (-3dB)

Misalignment, (1σ)

g-Sensetivity

Measurement Range

Bias (In Run Stability)(1σ)

Bias (Over Temperature) (1σ)

Noise Density

Scale Factor

Error over temp., (1σ)

Non-linearity

Bandwidth (-3dB)

Bias (In Run Stability)(1σ)

Bias (Over Temperature) (1σ)

Noise Density

Scale Factor

Error over temp., (1σ)

Non-linearity

Bandwidth (-3dB)

Temperature range (operating)

Input Power

Power Consumption (12V input)

Interfaces

Size

Weight

Vibration (20Hz - 2000Hz)

Out Data Rate

Input Power

Power Consumption (12V input)

Interfaces

Size

Weight

Vibration (20Hz - 2000Hz)

Out Data Rate

±2000 deg/sec

5 deg/hr

< 50 deg/hr

0.11 deg/sqrt(hr)

0.0017 deg/sec/sqrt(Hz)

0.03%

< 0.025% of FS

100-250 Hz

< 3 arc min

TBD

5 deg/hr

< 50 deg/hr

0.11 deg/sqrt(hr)

0.0017 deg/sec/sqrt(Hz)

0.03%

< 0.025% of FS

100-250 Hz

< 3 arc min

TBD

±15 g

10 ug

1 mg

30 ug/sqrt(Hz)

0.1%

< 0.01% of FS

100 - 250 Hz

10 ug

1 mg

30 ug/sqrt(Hz)

0.1%

< 0.01% of FS

100 - 250 Hz

-40C - +85C

4.5 V to 32 V

90 mW

UART, SPI

24 x 21.3 x 13,3 mm

< 10 gram

6 g RMS

250 - 5000 Hz

4.5 V to 32 V

90 mW

UART, SPI

24 x 21.3 x 13,3 mm

< 10 gram

6 g RMS

250 - 5000 Hz

For all IMUs and inertial modules, we offer special boards that allow you to conveniently and correctly mount and connect our units for evaluation purposes. The boards have more common connectors and they are a very simple solution when you need a quick start or your application allows a bit extra volume for the purpose. Please, refer to our Accessories section for more details.

MANUAL

SAMPLE DATA

downloads

* - best achievable specs, provided correct mounting, no significant temperature gradients or electro-magnetic interference